BA-88A MINDRAY PDF
Lab supply | First online laboratory shop, and selling where new features were introduced. practice. Instrumental factor must be checked. Chema Diagnostica will not be responsible for instrumental bad programming. Mindray BAA. Mindray) owns the intellectual property rights to this Mindray product and this manual. This manual may refer to information protected by copyrights or patents.
|Published (Last):||9 February 2011|
|PDF File Size:||15.89 Mb|
|ePub File Size:||8.28 Mb|
|Price:||Free* [*Free Regsitration Required]|
The supplier supports Trade Assurance — A free service that protects your orders from payment to delivery. Table of Contents Add to my manuals Add. Pinch the two buckles on the left and right side of the pump shell and detach the pump head carefully.
When replacing the bba-88a components, take care to protect the components.
MINDRAY BAA SERVICE MANUAL Pdf Download.
Preventing Electric Shock Please observe the following instructions to prevent electric shock. Date of Manufacture Manufacturer CE marking. Introduction of the Modules The hardware system of the BAA is divided into the following modules or boards: User-friendly interface and multi-language operation software. BA A is a Semi – Auto chemistry analyzer and a computer in one unit.
Call Send a quick message. The procedure for adjusting is shown as follows: Page 61 Click Lower upgr and the Lower computer upgrading dialog box will be displayed. NOTE The assembly or disassembly of the filter wheel, motor or home position sensor should be followed with home position deviation adjustment.
Page of 79 Go. Pinch both ends of the joint and press the buckle to disconnect mindrqy joint.
Please pay attention to the following: RSC Hard disk the disk on which the software is installed: After calibration is completed, calibration formula can not be gained, but the previous calibration formula can be used minvray calculate the formula and the test will not be affected.
Year of Establishment It has pop-up keypad. When maintaining the components on the side of analogue board and air duct, it is not necessary to dismount the flow cell base assembly and maintenance can be done on ba-88q dismounted photometer assembly.
BAA Semi-Auto Chemistry Analyzer at Rs /piece | Chemistry Analyzers | ID:
Relevancy Transaction Level Response Rate. BAA Semi automatic Mindray chemistry analyzer. Mindray 7inch touch screen medical biochemistry analyzer BAA Semi-Auto chemistry analyzer price for hospital ba-88x lab.
If the optical surface of the flow cell is contaminated, use cloth soaked with certain amount of absolute alcohol to clean it. Pull the wire out and you can see the wire joint connecting the lamp assembly, with one end connecting to the lamp and the other end connecting to the inside of the instrument.
Page 7 This manual is geared for service personnel authorized by Mindray.
BA-88A Semi-Auto Chemistry Analyzer
Troubleshooting Troubleshooting This chapter presents all warning messages and recommended corrective actions, which should be taken in time once any error occurs. Software Upgrading If the optical surface of the flow cell is contaminated, use cloth soaked with certain amount of absolute alcohol to clean it.
Before detected by control module, all input signals should go through the level translation circuit. More products from this supplier. To wipe off dust from the system surface, use a soft, clean and wet not mindrah wet cloth, soaked with mild soap solution if necessary, to clean the surface. Any person, who connects additional equipment to the signal input or output ports and configures an IVD system, is responsible for ensuring that the system work normally and complies with the safety and EMC The structure is shown as the following figure.
System Installation System Installation Basic Make sure the installation site of the hospital meet the system installation requirements on space, power supply and environment. Click Maintenance on the main screen of the mindrau. Page 18 Analogue Board Page 65 Locate the position where the peristaltic pump is installed.
If obvious blow-up is found, skewed installation of the lens might be the cause and the readjustment might be needed. Finally, it will be output at the inputting end of the AD converter through selectable switch. Page 58 It will take about 15 minutes for the system to initialize, then the interface to detect the upgrading program on the U disk will be displayed.
Mindray BA-88A Service Manual
Treating Waste Liquids Please ba-88a the following instructions to prevent environmental pollution and personal injury caused by waste. Got it, continue to print. It will take 10 minutes to upgrade the operating software.
Tubing should be checked irregularly and replaced in time. Microplate washer MWA Mindray. Install all the boards back and ensure mindra the connections are well connected;